Frequently Asked Questions About our Products and Solutions
Submit an enquiry
Submit an enquiry
(1992 total: showing 21-40)
Are the traditional methods of using RTD to measure the back-of-module temperature not used with the CS240 and CS240DM because these methods may cause hot spots?
Can multiple instances of Campbell Scientific SURVEYOR be run on a PC at the same time?
How tight do the wing-nuts on the CM226 integrated mounting stand need to be tightened to secure the solar sensor?
Can the SG000 determine the exact distance to a lightning strike?
When selecting a recalibration service for a pyranometer, how do the services differ and what is the importance of ISO 9060?
What are some distinguishing features of the MetSENS compact weather sensors?
Can the CSAT3BH heaters be controlled manually?
What happens to the data when the CSAT3BH heaters are on?
How often does the LevelVUE™B10 need to be serviced?
For the LevelVUE™B10, can I use the temperature value as ambient air temperature?
To get accurate readings with the LevelVUE™B10, is it necessary to know the temperature of the water?
Is it necessary to use a stilling well with the LevelVUE™B10?
Can the LevelVUE™B10 be used in groundwater applications?
Does the LevelVUE™B10 log its own data, or does it require a separate data logger?
Does the LevelVUE™B10 work with PLCs and other controllers that do not support SDI-12?
Does the LevelVUE™B10 work with data loggers from other manufacturers?
How often should the CR1000 be recalibrated?
When should a data acquisition (DAQ) system be used instead of a PC-based system?
How can the age of a sealed rechargeable battery be determined?
If the CS135 reports a warning or alarm, when should a visit to the sensor be made?