The CS655 is a multiparameter smart sensor that uses innovative techniques to monitor soil volumetric-water content, bulk electrical conductivity, and temperature. It outputs an SDI-12 signal that many of our dataloggers can measure. It has shorter rods than the CS650, for use in problem soils.
This product is supplied with a 3 m cable as standard, other lengths available to order.Read More
The CS655 consists of two 12-cm-long stainless steel rods connected to a printed circuit board. The circuit board is encapsulated in epoxy and a shielded cable is attached to the circuit board for data logger connection.
The CS655 measures propagation time, signal attenuation, and temperature. Dielectric permittivity, volumetric water content, and bulk electrical conductivity are then derived from these raw values.
Measured signal attenuation is used to correct for the loss effect on reflection detection and thus propagation time measurement. This loss-effect correction allows accurate water content measurements in soils with bulk EC ≤8 dS m-1 without performing a soil-specific calibration.
Soil bulk electrical conductivity is also calculated from the attenuation measurement. A thermistor in thermal contact with a probe rod near the epoxy surface measures temperature. Horizontal installation of the sensor provides accurate soil temperature measurement at the same depth as the water content. Temperature measurement in other orientations will be that of the region near the rod entrance into the epoxy body.
|Measurements Made||Soil electrical conductivity (EC), relative dielectric permittivity, volumetric water content, soil temperature|
|Required Equipment||Measurement system|
|Soil Suitability||Short rods are easy to install in hard soil. Suitable for soils with higher electrical conductivity.|
|Sensing Volume||3600 cm3 (~7.5 cm radius around each probe rod and 4.5 cm beyond the end of the rods)|
|Electromagnetic||CE compliant (Meets EN61326 requirements for protection against electrostatic discharge and surge.)|
|Operating Temperature Range||-50° to +70°C|
|Sensor Output||SDI-12; serial RS-232|
|Warm-up Time||3 s|
|Measurement Time||3 ms to measure; 600 ms to complete SDI-12 command|
|Power Supply Requirements||6 to 18 Vdc (Must be able to supply 45 mA @ 12 Vdc.)|
|Maximum Cable Length||610 m (2000 ft) combined length for up to 25 sensors connected to the same data logger control port|
|Rod Spacing||32 mm (1.3 in.)|
|Ingress Protection Rating||IP68|
|Rod Diameter||3.2 mm (0.13 in.)|
|Rod Length||120 mm (4.7 in.)|
|Probe Head Dimensions||85 x 63 x 18 mm (3.3 x 2.5 x 0.7 in.)|
|Cable Weight||35 g per m (0.38 oz per ft)|
|Probe Weight||240 g (8.5 oz) without cable|
|Active (3 ms)||
|Quiescent||135 µA typical (@ 12 Vdc)|
|Range for Solution EC||0 to 8 dS/m|
|Range for Bulk EC||0 to 8 dS/m|
|Accuracy||±(5% of reading + 0.05 dS/m)|
|Precision||0.5% of BEC|
Relative Dielectric Permittivity
|Range||1 to 81|
Volumetric Water Content
|Range||0 to 100% (with M4 command)|
|Water Content Accuracy||
|Range||-50° to +70°C|
Please note: The following shows notable compatibility information. It is not a comprehensive list of all compatible products.
External RF sources can affect the probe’s operation. Therefore, the probe should be located away from significant sources of RF such as ac power lines and motors.
Multiple CS655 probes can be installed within 4 inches of each other when using the standard data logger SDI-12 “M” command. The SDI-12 “M” command allows only one probe to be enabled at a time.
The CS650G makes inserting soil-water sensors easier in dense or rocky soils. This tool can be hammered into the soil with force that might damage the sensor if the CS650G was not used. It makes pilot holes into which the rods of the sensors can then be inserted.
Number of FAQs related to CS655: 55
Expand AllCollapse All
Yes. There is surge protection built into the sensor electronics. The sensor survives a surge of 2 kV at 42 ohm line-to-ground on digital I/O and 2 kV at 12 ohm line-to-ground on power. It also survives a surge of 2 kV at 2 ohm line-to-ground on the rods.
If additional surge protection is required, consider using the SVP100 Surge Voltage Protector DIN Rail with Mounting Hardware.
The CS650-series sensors have the same rugged epoxy and stainless-steel rods that have been used for water content reflectometers since the CS615-L model was introduced in 1995. There are CS615-L and CS616 sensors in many locations that have been in continuous use for more than ten years with no reported problems. If a CS650 or CS655 remains undamaged by external forces such as lightning, harsh chemicals, or animal actions, the sensor is expected to continue working for decades.
Campbell Scientific strongly discourages shortening the sensor’s rods. The electronics in the sensor head have been optimized to work with the 12 cm long rods. Shortening these rods will change the period average. Consequently, the equations in the firmware will become invalid and give inaccurate readings.
Yes, but the pots would have to be large. The CS650 and CS655 can detect water as far away as 10 cm (4 in.) from the rods. If the pot has a diameter smaller than 20 cm (8 in.), the sensor could potentially detect the air around the pot, which would underestimate the water content. In addition, potting soil is typically high in organic matter and clay, causing the probable need for a soil-specific calibration.
No. The abrupt permittivity change at the interface of air and saturated soil causes a different period average response than would occur with the more gradual permittivity change found when the sensor rods are completely inserted in the soil.
For example, if a CS650 or a CS655 was inserted halfway into a saturated soil with a volumetric water content of 0.4, the sensor would provide a different period average and permittivity reading than if the probe was fully inserted into the same soil when it had a volumetric water content of 0.2.
The CS650 has rods that are 30 cm long, and the CS655 has rods that are 12 cm long. The difference in rod length causes some changes in specifications. For example, the CS650 is slightly more accurate in its permittivity and water content readings, but the CS655 works over a larger range of electrical conductivity. In addition, the CS650 handles a larger measurement volume and provides good accuracy in low EC (electrical conductivity) sand and sandy loam. The CS655 is typically more accurate in soil, works well over a wide range of soil textures and EC, and is easier to install because of its shorter rods.
The volumetric water content reading is the average water content over the length of the sensor’s rods.
Probably not. The principle that makes these sensors work is that liquid water has a dielectric permittivity of close to 80, while soil solid particles have a dielectric permittivity of approximately 3 to 6. Because the permittivity of water is over an order of magnitude higher than that of soil solids, water content has a significant impact on the overall bulk dielectric permittivity of the soil. When the soil becomes very dry, that impact is minimized, and it becomes difficult for the sensor to detect small amounts of water. In air dry soil, there is residual water that does not respond to an electric field in the same way as it does when there is enough water to flow among soil pores. Residual water content can range from approximately 0.03 in coarse soils to approximately 0.25 in clay. In the natural environment, water contents below 0.05 indicate that the soil is as dry as it is likely to get. Very small changes in water content will likely cause a change in the sensor period average and permittivity readings, but, to interpret those changes, a very careful calibration using temperature compensation would need to be performed.
The electrical conductivity (EC) of sea water is approximately 48 dS/m. The CS655 can measure permittivity in water with EC between 0 and 8 dS/m. EC readings become extremely unstable at conductivities higher than 8 dS/m and are reported as NAN or 9999999. Because EC is part of the permittivity equation, an EC reading of NAN leads to a permittivity reading of NAN as well. Thus, the CS655 cannot provide good readings in sea water.
With regard to sea ice, the electrical conductivity drops significantly when sea water freezes and the permittivity changes from approximately 88 down to approximately 4, as the water changes from a liquid to a solid state. With both EC and permittivity falling to levels that are within the CS655 measurement range, the sensor is expected to give valid readings in sea ice. The sensor is rugged and can withstand the cold temperatures. However, as the ice melts, there will be a point at which the electrical conductivity becomes too high to acquire a valid reading for either permittivity or electrical conductivity.