The CS650 is a multiparameter smart sensor that uses innovative techniques to monitor soil volumetric water content, bulk electrical conductivity, and temperature. It outputs an SDI-12 signal that many of our dataloggers can measure.
This product is supplied with a 3 m cable as standard, other lengths available to order.Read More
The CS650 consists of two 30-cm-long stainless steel rods connected to a printed circuit board. The circuit board is encapsulated in epoxy and a shielded cable is attached to the circuit board for datalogger connection.
The CS650 measures propagation time, signal attenuation, and temperature. Dielectric permittivity, volumetric water content, and bulk electrical conductivity are then derived from these raw values.
Measured signal attenuation is used to correct for the loss effect on reflection detection and thus propagation time measurement. This loss-effect correction allows accurate water content measurements in soils with bulk EC ≤3 dS m-1 without performing a soil specific calibration.
Soil bulk electrical conductivity is also calculated from the attenuation measurement. A thermistor in thermal contact with a probe rod near the epoxy surface measures temperature. Horizontal installation of the sensor provides accurate soil temperature measurement at the same depth as the water content. Temperature measurement in other orientations will be that of the region near the rod entrance into the epoxy body.
|Measurements Made||Soil electrical conductivity (EC), relative dielectric permittivity, volumetric water content (VWC), soil temperature|
|Required Equipment||Measurement system|
|Soil Suitability||Long rods with large sensing volume (> 6 L) are suitable for soils with low to moderate electrical conductivity.|
|Sensing Volume||7800 cm3 (~7.5 cm radius around each probe rod and 4.5 cm beyond the end of the rods)|
Meets EN61326 requirements for protection against electrostatic discharge and surge.
|Operating Temperature Range||-50° to +70°C|
|Sensor Output||SDI-12; serial RS-232|
|Warm-up Time||3 s|
|Measurement Time||3 ms to measure; 600 ms to complete SDI-12 command|
|Power Supply Requirements||6 to 18 Vdc (Must be able to supply 45 mA @ 12 Vdc.)|
|Maximum Cable Length||610 m (2000 ft) combined length for up to 25 sensors connected to the same data logger control port|
|Rod Spacing||32 mm (1.3 in.)|
|Ingress Protection Rating||IP68|
|Rod Diameter||3.2 mm (0.13 in.)|
|Rod Length||300 mm (11.8 in.)|
|Probe Head Dimensions||85 x 63 x 18 mm (3.3 x 2.5 x 0.7 in.)|
|Cable Weight||35 g per m (0.38 oz per ft)|
|Probe Weight||280 g (9.9 oz) without cable|
|Active (3 ms)||
|Quiescent||135 µA typical (@ 12 Vdc)|
|Range for Solution EC||0 to 3 dS/m|
|Range for Bulk EC||0 to 3 dS/m|
|Accuracy||±(5% of reading + 0.05 dS/m)|
|Precision||0.5% of BEC|
Relative Dielectric Permittivity
|Range||1 to 81|
Volumetric Water Content
|Range||0 to 100% (with M4 command)|
|Water Content Accuracy||
|Range||-50° to +70°C|
Please note: The following shows notable compatibility information. It is not a comprehensive list of all compatible products.
External RF sources can affect the probe’s operation. Therefore, the probe should be located away from significant sources of RF such as ac power lines and motors.
Multiple CS650 sensors can be installed within 4 inches of each other when using the standard datalogger SDI-12 “M” command. The SDI-12 “M” command allows only one probe to be enabled at a time.
The CS650G makes inserting soil-water sensors easier in dense or rocky soils. This tool can be hammered into the soil with force that might damage the sensor if the CS650G were not used. It makes pilot holes into which the rods of the sensors can then be inserted.
Number of FAQs related to CS650: 54
Expand AllCollapse All
Campbell Scientific strongly discourages shortening the sensor’s rods. The electronics in the sensor head have been optimized to work with the 30 cm long rods. Shortening these rods will change the period average. Consequently, the equations in the firmware will become invalid and give inaccurate readings.
Mine tailings are highly corrosive and have high electrical conductivity. Some customers have successfully used water content reflectometers, such as the CS650 or the CS655, to measure water content in mine tailings by coating the sensor rods with heat-shrink tubing. This affects the sensor output, and a soil-specific calibration must be performed. Care must be taken during installation to avoid damaging the heat-shrink tubing and exposing the sensor’s rods. In addition, covering the sensor’s rods invalidates the bulk electrical conductivity reading. Unless the temperature reading provided by the CS650 or the CS655 is necessary, a better option may be to use a CS616 with coated rods.
The CS650 has rods that are 30 cm long, and the CS655 has rods that are 12 cm long. The difference in rod length causes some changes in specifications. For example, the CS650 is slightly more accurate in its permittivity and water content readings, but the CS655 works over a larger range of electrical conductivity. In addition, the CS650 handles a larger measurement volume and provides good accuracy in low EC (electrical conductivity) sand and sandy loam. The CS655 is typically more accurate in soil, works well over a wide range of soil textures and EC, and is easier to install because of its shorter rods.
A CS650 or CS655 can be ordered with an SDI-12 address option of -VS. With the -VS option, the SDI-12 address is set at the factory before the sensor is shipped. The last digit of the sensor’s serial number becomes that sensor’s SDI-12 address. Typically, the -VS option is chosen when there are multiple sensors that will communicate with the data logger on the same SDI-12 communications terminal.
If the -VS option is not selected when ordering, the CS650 or CS655 will ship with its SDI-12 address set to 0 (the default -DS option). The address can be changed to a non-zero value using the A200 Sensor to PC Interface or by connecting the sensor to an SDI-12 communications terminal and sending the aAb! Command as described in the “SDI-12 Sensor Support” appendix of the CS650/CS655 manual.
The volumetric water content reading is the average water content over the length of the sensor’s rods.
The bulk electrical conductivity (EC) measurement is made along the sensor rods, and it is an average reading of EC over that distance at whatever depth the rods are placed.
The cable properties and power requirements of the CS650 and the CS655 are such that communication with a data logger may work for cable lengths greater than 2,000 ft. If multiple sensors are communicating through the same universal or control terminal, the total length of all of those sensors must not exceed 2,000 ft.
In practice, it is less expensive to purchase a new data logger than to buy a CS650 or CS655 with 2,000 ft of cable. If the cable is run through conduit, or if a 2,000 ft long trench needs to be excavated, then the installation cost becomes more expensive than buying another data acquisition system and sensors with shorter cables.
Modifications to the CS650 or CS655, including shortening the cable, will void the warranty. However, shortening the cable will not affect the sensor’s performance. If a decision is made to shorten the cable, care should be taken to avoid damaging the cable jacket and exposing bare wire except at the ends that connect to the data logger or multiplexer terminals.
No. The principle that makes these sensors work is that liquid water has a dielectric permittivity of close to 80, while soil solid particles have a dielectric permittivity of approximately 3 to 6. Gasoline and other hydrocarbons have dielectric permittivities in the same range as soil particles, which essentially make them invisible to the CS650 and the CS655.
The permittivity of saturated sediments in a stream bed is expected to read somewhere between 25 and 42, while the permittivity of water is close to 80. A CS650 or CS655 installed in saturated sediments could be used to monitor sediment erosion. If the permittivity continuously increases beyond the initial saturated reading, this is an indication that sediment around the sensor rods has eroded and been replaced with water. A calibration could be performed that relates permittivity to the depth of the rods still in the sediment.